Since 95% of pipette failures are sealing system-related, cleaning and inspecting your pipette regularly will go a long way toward assuring solid performance.
Clean the shaft, seal/o-ring and piston with de-ionized water
Use solutions designed to remove DNA & RNA when appropriate
Replace the seal and o-ring if cracks or signs of leakage occur
Avoid using corrosive solvents and clean the pipette immediately when you do
Hanging pipettes when not in use will assure liquids completely drain out
A simple performance test known as "quick check" is a great way to evaluate your ability and the accuracy and precision of your pipettes. By periodically assessing whether or not your pipettes are performing within tolerances, you can assure data integrity and reduce your overall risk.
For detailed instructions and tools to help you do a quick check, visit Quick Check.
When choosing a pipette service provider:
Insist that original manufacturer's parts are used for replacements.
Balances should be high quality, certified regularly and appropriate for the volume of the pipette being tested or calibrated (e.g., the smaller the volume, the greater the sensitivity required).
Technicians should be trained and routinely tested; calibration technicians must be certified.
The quality of the work is guaranteed. A provider who includes a warranty is more likely to perform quality work and guarantees that it is done correctly.
Questions? Just ask!
METTLER TOLEDO offers pipette support through its local representatives as well as through its technical support team.
Thank you for visiting www.mt.com. We have tried to optimize your experience while on the site, but we noticed that you are using an older version of a web browser. We would like to let you know that some features on the site may not be available or may not work as nicely as they would on a newer browser version. If you would like to take full advantage of the site, please update your web browser to help improve your experience while browsing www.mt.com.