Turbidity Meter / Turbidity Sensor Frequently Asked Questions (FAQ)
What is turbidity?
Turbidity is an optical characteristic which refers to the degree of clarity of a liquid. Turbidity in water i...
What is turbidity?
Turbidity is an optical characteristic which refers to the degree of clarity of a liquid. Turbidity in water is caused by individual suspended particles or colloidal matter that scatters or obstructs light transmittance: the higher the concentration of suspended particles/colloidal matter, the higher the turbidity. Such particles are normally too small to be detected by the human eye; therefore, turbidity measurement has to be done with a turbidity meter. Everyday examples of turbid liquids are:
Why is turbidity measurement important?
In many cases, turbidity meters are used for the evaluation of the quality of water or the efficiency of a filtration process. Turbidity measurements are important indicators in many industries and applications as they not only influence the yield of the industrial process but also detect factors which are detrimental to a system.
What influences turbidity measurements?
As mentioned above, turbidity measurement determines the degree that suspended particles in a liquid medium, scatter light. The scattering depends on
As a consequence of the three points above, it is only possible for turbidity to be used as a characteristic property of a sample if the measurement method is standardized. For example, in many brewery applications the liquid to be measured is yellowish in color and contains yeast particles. Therefore, to check for filter breakthrough, the amount of forward and side-scattered light is measured at an angle of 25o and 90o compared to the light source for quality assurance purposes. Red (650 nm) and blue (460 nm) light sources are also specified in guidelines for turbidity and color monitoring.
What are the common interferences in turbidity measurement?
Turbidity measurement can be adversely affected by the presence of light-absorbing substance or those that fluoresce. This interference can be minimized by using wavelengths other than those in visible light (i.e. 860 nm near-infrared).
Air bubbles can interfere with turbidity measurement, so choosing an installation point where the presence of air bubbles is minimal, can be important.
Windows fouling is also critical in turbidity measurement. State of-the-art technology and the ratio principle can compensate for sample color changes, aging of the lamp used as the light source and optical window fouling.
What is a turbidity meter?
A turbidity meter is a sensor that measures turbidity. Turbidity sensors have two things in common:
What are the common turbidity meter measurement methods in the market?
There are several types of in-line turbidity sensors that are optimized for specific measurement ranges and different applications.
How do you calibrate a turbidity meter?
Calibrating a turbidity meter can be done in three different ways with a transmitter. The first type is called manual calibration. This is the fastest but lowest level of calibration. The user can change the offset and the slope, and a measuring value is calculated and displayed.
The second type is called multi-point calibration. This type of calibration achieves the best linearity for the process being measured. Two, three, four and five point calibrations can be performed – typically off-line with an uninstalled sensor.
The third type is process or in situ calibration. This is an on-line calibration where the operator enters the process calibration menu on the transmitter and saves the current turbidity reading. The operator then takes a sample of the process liquid to measure it against a laboratory instrument in order to obtain a reference turbidity measurement. Depending on the turbidity sensor and transmitter, multiple point in situ calibrations are possible (e.g. METTLER TOLEDO M800 transmitter and backscattered light turbidity sensor).
METTLER TOLEDO turbidity meter/turbidity sensors can be used in many industries such as:
How do you select the right turbidity sensor?
The turbidity sensor should be selected based on the required measurement range as well as the application. METTLER TOLEDO offers several versatile turbidity sensors that meet the requirements of many different applications, such as:
Please contact your local METTLER TOLEDO representative for details.
Turbidity and color measurement for upstream and downstream production in the brewery and beverage industries, using scattered light and absorption method.
Broad measurement range from medium to high turbidity - quick process adaption and easy installation.
Rugged and easy to use turbidity system designed for use in phase separation by turbidity/color and for product identification processes.
Accurate wort and finished product control
12-point calibration for precise measurement
Sensor monitoring for measurement reliability
Easy installation; fits in sight-glass
Reduce your investment with unique upstream model
Suitable for pharmaceutical, chemical or wastewater applications
In situ multi-point calibration
Small 12mm diameter sensor saves valuable space
Pg 13.5 thread for easy installation and interface into housings
Reliable solution for food & beverage industry
Simple and cost effective
measurement with direct 4/20 mA output
Various sensor options available for special applications
Support & Repair
Maintenance & Optimization
Calibration & Quality
Training & Consulting