The accuracy of moisture analyzer results may be influenced by several factors. The most important is variability of the heating temperature. Routine testing of the heating and weighing unit are therefore necessary to ensure reliable determination of the moisture content and correct functioning of the instrument.
SmartCal™ is an innovative test substance that offers a fast and efficient way to verify moisture analyzer performance. Testing your moisture analyzer regularly with SmartCal™ ensures that your measurements are reliable and fulfil the demands of your industry’s regulations.
The cSmartCal™ test substance is certified by the independent German Federal Institute for Materials Research and Testing. The Certificate of Analysis satisfies the strictest regulatory requirements.
Simply select the integrated SmartCal™ method on the instrument, distribute the contents of the test sachet evenly over the sample pan and start the measurement. After ten minutes, you know if your instrument’s performance is satisfactory or not.
SmartCal™ measurement reports make it easy to monitor and interpret your instrument’s performance over time. Results can be stored directly in the instrument or manually entered into our free validated Excel report sheet.
With our easy to use temperature kit you can instantly verify the operating temperature of the heating unit and adjust the instrument to ensure reliable results.
This white paper describes the influences and sources of error which may be present when conducting moisture analyses. It discusses the routine tests which are necessary to ensure reliable determination of the moisture content and correct functioning of the instrument.
Read the paper and discover the benefits of a fast and convenient interim performance test using SmartCal. The publication describes how this method relates to the traditional routine testing of moisture analyzers.