# Conductivity Sensor | Conductivity Probe

## Conductivity Sensors for Process and Pure Water Measurement

How can we assist you?

## Conductivity Sensor and Conductivity Analyzer - Frequently Asked Questions (FAQ)

What is conductivity Electrical conductivity is the ability of a material to carry an electrical current. It is generally measured with an elect...

What is conductivity

Electrical conductivity is the ability of a material to carry an electrical current. It is generally measured with an electrical conductivity meter (EC Meter) or conductivity analyzer.

What is resistivity?

Electrical resistivity is the reciprocal of conductivity. Resistivity is the intrinsic property that quantifies how strongly a given material opposes the flow of electric current.

Why do we measure conductivity?

Electrical conductivity has been measured using EC meters for many years and it is still an important and widely used analytical parameter today. Electrical Conductivity Meters offer an easy and economical way to provide an indication of the purity of the measured medium, typically, water (the higher the conductivity reading, the higher is the concentration of dissolved ions in the water). The excellent reliability, sensitivity, response and relatively low cost of a conductivity analyzer makes conductivity a valuable, easy-to-use tool for quality control. In some applications, the purity measurement is made as resistivity (the reciprocal of conductivity).

What does a conductivity sensor measure?

A conductivity sensor (or conductivity probe) measures the ability of a solution to conduct an electrical current. It is the presence of ions in a solution that allow the solution to be conductive: the greater the concentration of ions, the greater the conductivity. You can find more information about METTLER TOLEDO in-line conductivity electrodes:

What is the measuring principle of a conductivity sensor?

A conductivity probe consists of an electrode pair, to which a voltage is applied. The conductivity sensor measures the flowing current and calculates the conductivity.

What unit is conductivity measured in?

Conductivity is measured in siemens per cm (S/cm). A conductivity of 1 S/cm is actually quite high, so most conductivity measurements involve solutions where conductivity is measured in mS/cm (thousandths of a S/cm) or in μS/cm (millionths of a S/cm).

How many types of conductivity sensors are there?

There are three technology types of conductivity probe used for process conductivity measurement in an in-line EC meter:

How does a 2-electrode conductivity meter work?

The classical 2-electrode conductivity meter consists of two parallel plates used for process conductivity measurements. An alternating current voltage is applied across the two electrodes, and the resistance between them is measured. The 2-electrode conductivity probe is used for water conditioning and purification stages where it is capable of detecting minimal levels of impurities in ultrapure water.

How does a 4-electrode conductivity meter work?

The 4-electrode EC meter works with an additional electrode pair. The outer electrodes are the current electrodes to which alternating current is applied; they are driven in the same manner as the 2-electrode conductivity sensor. The in-line conductivity electrode measures are placed in the electric field of the current electrodes and measure the voltage with a high impedance amplifier. The current flowing through the outer electrodes and the solution can be accurately measured by the circuit. If the voltage across the inner electrodes and the current are known, the resistance and conductance can be calculated. The advantage of the 4-electrode conductivity sensor lies in the fact that there is negligible current flowing through the inner electrodes where the measurement is made. Therefore, no polarization effects occur which would otherwise influence the measurement. The 4-electrode conductivity sensor is also less sensitive to measuring errors through electrode fouling. The 4-electrode sensors are for middle to high ranges.

How does an inductive conductivity probe work?

The METTLER TOLEDO inductive conductivity probe is constructed like a pair of transformer coils where the solution to be measured is the core of the transformer. The parallel coils are closely spaced and embedded within a polymer body like a donut which is immersed into the solution to ensure accurate process conductivity measurement. There are no electrodes and usually no metal contacting the solution. One coil is energized with alternating current and the signal induced into the second coil is related to the conductivity of the solution flowing through and around the sensor. The cell constant is determined by the diameter of the hole, among other factors. The inductive conductivity sensor covers mid to very high conductivity ranges, and is particularly resistant to fouling.  Because it is non–contact, it is particularly suitable to be used in chemical (corrosive) applications where metal electrodes could be damaged by the media.

What does cell constant mean?

The cell constant is the ratio of the distance between the electrodes of a conductivity probe to the area of the electrodes in 2- and 4-electrode conductivity sensors. The smaller the cell constant, the more precise the in-line conductivity electrode will be in determining changes of the conductivity in media. However, a small cell constant reduces a sensor's measuring range. Accurate conductivity measurement requires an accurate measurement of the cell constant, which is determined by calibration. For METTLER TOLEDO conductivity sensors, the cell constant is precisely measured and is documented on each sensor's Quality Certificate. Calibration solutions can be traced back to the National Institute of Standards and Technology (NIST).

How do you calibrate a conductivity sensor?

A  METTLER TOLEDO conductivity sensor used in an on-line EC Meter can be calibrated against a solution of known conductivity (much like calibrating a pH sensor against a solution of a known pH). Alternatively, a device can be used that contains a range of very precise resistors that duplicate known conductivity measurements.

When do you need to perform a conductivity sensor calibration or verification?

In general, the cell constant of the sensor will not change; however, if the sensing elements are altered in some way (e.g. solid deposition or other fouling of the electrodes or insulator of the sensor, loss of electrode material through corrosion) the cell constant will change. METTLER TOLEDO conductivity sensors are factory calibrated and the cell constant is precisely determined. Therefore, calibration is typically not required while using the in-line conductivity sensor. However, it is recommended to verify the sensor or make calibration adjustment if necessary on an annual basis. The frequency of verification or calibration is very much dependent on the applications or on the plant standard operating procedure requirements.

Does temperature affect the conductivity measurement?

Conductivity is strongly temperature dependent. As the temperature of a sample increases, the viscosity of the sample decreases which leads to increased mobility of the ions. Therefore, the observed conductivity of the sample also increases even though the ion concentrations may remain constant.

In good practices, every conductivity sensor result must be specified with a temperature or be temperature compensated, usually to the industry standard of 25 degrees Celsius. Because temperature is also dependent to various samples, the appropriate temperature compensation algorithms must be chosen carefully.

### METTLER TOLEDO - Thornton Conductivity / Resistivity Sensors for Pure Water to Sea Water Applications

Sensors with 2 or 4 electrodes for measuring conductivity across a wide range of industrial applications including pharmaceutical, chemical and wastewater.
Easy-to-use inductive conductivity sensors provide accurate measurement without electrodes that are in contact with the process medium.
Monitor pure water conductivity with an array of conductivity / resistivity sensors and degassed cation conductivity panels.
Built to withstand CIP & SIP processes
4-electrode models resist fouling in production
Ensure compliance through certification package
Plug and Measure functionality on ISM sensors
Resistant to fouling in chemical applications
Long lifetime due to strong chemical resistance
Individually tested to ensure accuracy
Approved for hazardous area use
High accuracy for pure water measurement
Monitor your full range of pharmaceutical waters
Detect corrosive contaminants in power plant water
Monitor for minute levels of impurities in UPW

Sensors with 2 or 4 electrodes for measuring conductivity across a wide range of industrial applications including pharmaceutical, chemical and wastewater.

Easy-to-use inductive conductivity sensors provide accurate measurement without electrodes that are in contact with the process medium.

Monitor pure water conductivity with an array of conductivity / resistivity sensors and degassed cation conductivity panels.

### Services

1300 659 761
Call Service
Uptime
Support & Repair
Compliance
Calibration & Quality
Expertise
Training & Consulting
Thank you for visiting www.mt.com. We have tried to optimize your experience while on the site, but we noticed that you are using an older version of a web browser. We would like to let you know that some features on the site may not be available or may not work as nicely as they would on a newer browser version. If you would like to take full advantage of the site, please update your web browser to help improve your experience while browsing www.mt.com.